It’s time to put your FieldKit to work and start recording environmental data out in the world.
Take your station to your deployment location, choose a data capture frequency, and document the process. The more information you gather in photographs and field notes during deployment, the more context you and your team have to work with when analyzing the data later.
Note: We believe that documentation is important. So while early versions of FieldKit will automatically start recording data, we’re working on new functionality to encourage users to realize the benefits of deploying in order to start recording data. That’s how this Product Guide is written.
1. Mount the Station Securely
Mount the case to a post or a tree with non-invasive mounting solutions, so that your station is protected from the elements and any curious animals!
2. Deploy Station
Once your station is secured, make sure that external instruments like probes or the weather cluster are properly set up (see Care Instructions and the sensor-specific deployment guidance below).
Ensure your power sources are in place (battery, micro-USB, or solar). See the Set Up Solar section for guidance on setting up your solar panel if that applies to you.
3. Name Deployment Location
In the app, name your location for future reference. Depending on your connectivity, we’ll detect your GPS coordinates or locate your position on a map.
4. Set Data Capture Schedule
Check the FieldKit app to ensure each sensor is still reading live data. These data readings are not being recorded at this stage. Your FieldKit station won’t be recording data until you complete the deployment process and hit “Record Data.”
Set your preferred data capture schedule. Indicate when and at what interval data readings should be recorded. Note that more frequent intervals of data capture drain the battery faster.
Choose which type of data capture schedule to set:
Simple Data Capture Schedule
Set your station to take data readings at a set interval of minutes or hours. This is useful for projects that need a consistent amount of data over a 24-hour period.
Example: Every 5 minutes
Example: Every 1 hour
Complex Data Capture Schedule
Set your station to take data readings at one or multiple specific times of the day, each at a set interval of minutes or hours. This may be useful for projects where most data is best gathered around certain events like sunrise or sunset, thus you’ll save battery by not recording (at all or at the same rate) around the clock.
Capture Time 1: Starts at 04:00 and ends at 07:00 and records every 1 minute
Capture Time 2: Starts at 07:00 and ends at 17:00 and records every 1 hour
Capture Time 3: Starts at 17:00 and ends at 19:00 and records every 1 minute
Capture Time 4: Starts at 19:00 and ends at 04:00 and records every 1 hour
5. Document Deployment Conditions with Field Notes
By providing some context to your deployment, not only are you helping the wider community better understand their environment—you’re also helping your future self and teammates gain clearer insights from the data.
Plus, if your station encounters issues, a little admin work now could save you lots of time and headache further down the road, especially if your FieldKit Station is deployed in a remote location.
Start with some notes. If your hands are full, you can take an audio recording of yourself through the app to capture your thoughts in the moment. Otherwise, type detailed notes that can inform data analysis and future visits to the station.
Then snap some pictures of your deployed FieldKit Station and the surrounding area. Is it set up near any landmarks? Are there any hazards around?
6. Hit “Record Data” to Begin Data Capture
Double-check the ‘Deployment Review’ page. Is your information correct? Review everything as if you are a future version of yourself or someone new to the study. Would you be able to understand everything? Great! You’re ready to start recording data. Hit “Record Data.”
7. Congratulations! Now Capture Some Initial Data
You are now deployed and recording data. We recommend hanging around to gather a few readings to sync that initial data capture to the FieldKit app. This will verify that your FieldKit is recording data properly.
8. Final Punch List Before Walking Away
Important: Take a moment to revisit our Care Instructions, the sensor-specific deployment guidance below and your Pre-Deployment Checklist to ensure you’ve set yourself up for success before you walk away. Do not skip this step.
Make sure that the gaskets in your FieldKit Case (lid and cable plate) are sitting snug in the grooves, nice and flat, not stretched or twisted, and the lid is properly closed using the lid clasps for maximum water resistance.
Clear up after yourself, leave the location as you found it, and leave with the peace of mind that your station is secured, protected and working.
Here are some things to think about when deploying Weather sensors.
Make Sure Nothing Has Shifted in Transit
Make sure your Rain Gauge is clear of the Anemometer and Wind Vane, and didn’t shift during assembly.
Mount 5M Above Ground
Ensure the Weather Instrument Cluster is set up in an area with unobstructed air flow and attached to a mast more than 5 m above the ground. This is to avoid boundary effects and record accurate measurements.
Is Everything Standing Straight?
Ensure the whole thing is plumb (standing up straight), you’ll tie a string with a weight on it to the top of the Wind Vane arm, then adjust the mast until it’s parallel with the string. Or, simply place a bubble level on top of the Rain Gauge.
Line Up with Due North
Locate the cardinal directions on the Wind Vane – you’ll place it in the field with the “N” mark facing north.
Anchor Securely in Case of Strong Winds
Anchor it securely in place when leaving it in the field, especially if your location might experience intense winds.
Here are some things to think about when deploying Water sensors.
Submerge the Probes
Ensure that the bottom of each probe is submerged in the water by at least 5 cm (~2 inches).
Protect Against Freezing
Make certain that pH, Electrical Conductivity, and Dissolved Oxygen probes are not allowed to freeze in place, as this will damage them. The temperature probe will be fine in freezing conditions.
Take Initial Measurement
For measurement of stage height in bodies of water with the distance sensor, make certain to first get a measurement of the height of the sensor above the bed directly below it using a plumb bob in order to know what distance to subtract from to get your stage height.
Consider Placing Probes in Stilling Well
If deploying in moving water, consider putting your probes in what is known as a stilling well: a section of pipe held above the bed to allow for water to slow down and allow sufficient contact time with probes to make an accurate measurement.
Use Protective Cable Cover
Use some sort of cable cover (such as wire loom or metal cable jacketing) to prevent the cables from being eaten by rodents or other wildlife.
Attach Probe Using Float if Necessary
If deploying in water with a changing level, probes may be attached so that they’re entirely submerged, but Dissolved Oxygen is measured with respect to overpressure, which is the sum of atmospheric pressure and the pressure of the water column. In order to keep this as simple as possible, we recommend putting this probe on a float inside of a stilling well so that it’s always submerged by the same amount. A simple float can be made from open cell poly packing foam and electrical tape.
Here are some things to think about when deploying solar panels.
Important: Before inserting the battery, solar and button cables, double check that you are connecting them to the correct sockets (labelled “BATTERY”, “SOLAR” and “BTN”). Inserting cables into the wrong sockets can permanently damage your FieldKit.
Optimize Sun Exposure for Solar Panels
Make certain that your solar panels are facing wherever they’ll be getting the most sun, and at an angle commensurate with your latitude. For example, if I’m at 42° N and I have trees screening to my east, I’ll aim my panel to the southwest at 42° altitude. If I were at 31° S, with trees screening to my northwest, I’d aim my panel northeast at 31° altitude.
Unsure About Anything?
For more ideas and help, ask in the FieldKit Community Forum.